It is not uncommon to find slip ratings on flooring products such as tiles and laminate. These are usually represented as R-values or Pendulum Test Values (PTV). Whilst many flooring professionals rely on R-values or PTV as a measure of slip resistance, these are commonly misunderstood. Following guidance by the Health & Safety Executive (HSE) and UK Slip Resistance Group (UKSRG), use of the Pendulum Tester is becoming increasingly prevalent due to its portability, cost-effectiveness and suitability on a wider range of flooring surfaces.
In this post we take a closer look at the Ramp test and Pendulum testing procedures, applications, benefits and disadvantages.
What is a Ramp Test?
The Ramp test involves an operator who is protected by a fall restraint harness. The operator, wearing standard footwear or barefoot (depending on the standard), walks backwards and forwards over a sample of a flooring material that has been evenly coated with a contaminant, Oil or soapy water depending on the standard. The Test starts with the ramp in a horizontal position; the operator gradually increases the angle of inclination until the limit of safe walking is reached and the test person “slips”. The acceptance angle obtained is used to express the degree of slip resistance (“R” value). The German ramp Test method is recognised worldwide (Europe, Australia, New Zealand, Singapore and the Ceramic Tile Institute of America (CTIOA)).
DIN 51130: A German standard that uses a cleated work boot and oil. A common problem is interpreting the data produced. It is presented as a scale running from R9 (the most slippery) to R13 (the least slippery). It is a common misconception that the scale starts at R1.
DIN 51097: A German standard that tests slip resistance of flooring in barefoot conditions using soapy water as a contaminant. Results are presented in a scale running from A (most slippery) to C (least slippery).
HSL ramp test: The HSL ramp test forms the basis for the British Standard (BS 4592) and uses footwear soled with slider 96 rubber and clean water as a contaminant. Results are presented as CoF (co-efficient of friction). A version of this test can be used to test footwear.
Also known as the British Pendulum Tester, the Portable Skid Resistance Tester was originally designed in the 1940s to measure the slip resistance of floors in government buildings. During the late 1950s, the instrument was adopted and redesigned to study problems in the design and maintenance of public highways, and to test the frictional resistance of new roads, road markings and iron works.
Today, the Portable Skid Resistance Tester is regularly used to test the slip resistance of roads, pedestrian walkways, office surfaces, shopping malls floors, factories, airports and sports facilities – both at the design stage, after installation and to investigate accidents.
Tests are carried out in accordance with a variety of different standards. These include: BS EN 13036-4, BS 7976-2, BS EN 1097-8, BS 812-114, ASTM E303, AS 4586 and AS 4663. The contaminant is commonly water and tests can be carried out In-situ.
Benefits of R Numbers:
Disadvantages of R values and ramp testing:
Benefits of Pendulum Skid Resistance tester and PTV (Pendulum tester Values):
Disadvantages of PTV and Pendulum Skid Testing:
Which is more cost effective?
The ramp test is a large fixed piece of equipment, used in a laboratory. It is a great piece of equipment that can be used to provide scientific information regarding how different types of flooring surfaces could perform with different types of footwear. ramp testing requires a sample of the flooring material to be tested in a laboratory setting. Therefore, ramp testing can typically only be used before a floor is laid,
Meanwhile, the Pendulum Tester is portable, and can be transported to carry out tests on-site with little disruption. This means pendulum slip tests are ideal for testing a variety of your premises’ existing flooring without the need to remove a sample to send it off for testing. The Pendulum can also be used to carry out internal product sample testing which can assist with R&D and offering better customer recommendations.
Overall the Pendulum tends to be much more cost effective. Due to the Ramp Test size, controlled laboratory conditions, and needed trained operators, the ramp test can often cost considerably more. Pendulum tests can also be the quicker option meaning they are ideal in legal situations where you need the results delivered promptly. Also, the HSE, UKSRG and TTA support, and recommend the Pendulum test method.
Munro instruments is a manufacturer of the Pendulum. If you would like further information on the Portable Skid Resistance Tester or Training, please get in touch: email (firstname.lastname@example.org) phone (+44 (0) 20 8551 7000).